- Open Access
A cheap and open HIV viral load technique applicable in routine analysis in a resource limited setting with a wide HIV genetic diversity
Virology Journal volume 14, Article number: 224 (2017)
HIV infection in Cameroon is characterized by a great viral diversity with all HIV-1 groups (M, N, O, and P) and HIV-2 in circulation. HIV group determination is very important if tailored viral load analysis and treatments are to be applied. In our laboratory, HIV viral load is carried out using two platforms; Biocentric and Abbott depending on the HIV group identified. Biocentric which quantifies HIV-1 group M is a cheap and open system useful in resource limited settings. The objective of this study was to compare the viral load analyses of serologically group-indeterminate HIV samples using the two platforms with the view of reducing cost.
Consecutive samples received between March and May 2014, and between August and September 2014 in our laboratory for HIV viral load analysis were included. All these samples were analyzed for their HIV groups using an in-house ELISA serotyping test. All HIV-1 group M samples were quantified using the Biocentric test while all other known atypical samples (HIV-1 groups N, O and P) were analyzed using the Abbott technique. HIV group-indeterminate samples (by serotyping) were quantified with both techniques.
Among the 6355 plasma samples received, HIV-1 group M was identified in 6026 (94.82%) cases; HIV-1 group O, in 20 (0.31%); HIV-1 group M + O, in 3 (0.05%) and HIV-2, in 3 (0.05%) case. HIV-group indeterminate samples represented about 4.76% (303/6355) and only 231 of them were available for analysis by Abbott Real-Time HIV-1 and Generic HIV Viral Load techniques. Results showed that 188 (81.39%) samples had undetectable viral load in both techniques. All the detectable samples showed high viral load, with a mean of 4.5 log copies/ml (range 2.1–6.5) for Abbott Real-Time and 4.5 log copies/ml (range 2–6.4) for Generic HIV Viral Load. The mean viral load difference between the two techniques was 0.03 log10 copies/ml and a good correlation was obtained (r 2 = 0.89; P < 0.001).
Our results suggest that cheaper and open techniques such as Biocentric could be useful alternatives for HIV viral load follow-up quantification in resource limited settings like Cameroon; even with its high viral diversity.
Human immunodeficiency virus (HIV) infection is a major public health problem in the world, particularly in sub-Saharan Africa where the majority of patients live. An outstanding characteristic of the virus is its genetic variability which has been attributed to high rates of mutation , recombination and viral turnover . To date, HIV is divided into two types: HIV-1 and HIV-2. HIV-1 has been subdivided into four phylogenetically distinct groups: M for major (or main), O for outlier, N for non-M/non-O (or new) and P [3, 4], while HIV-2 is subdivided into nine groups: A-I . According to the demographic health survey (DHS) 2011, the seroprevalence of HIV is estimated at 4.3% in Cameroon . This infection is marked by a great genetic diversity with the co-circulation of all types and groups (HIV-1 M-P and HIV-2). This diversity has been shown to impact on the diagnosis (possibility of false negative results) , on treatment; (some studies described that HIV-1 O were naturally resistant to the non-nucleoside reverse transcriptase inhibitor because of the presence of Y181C mutation in the RT gene ) and on follow-up of patients. Therefore, diagnostic techniques (screening and molecular biology), follow-up and treatment options are really dependent on the type and/or group of HIV. HIV-1 O for example has a broader diversity than HIV-1 M  and causes great difficulty for diagnosis. The high genetic diversity of HIV-1 also has a major impact on the plasma quantification of HIV-1 RNA [10, 11]. HIV group determination is very important if tailored viral load analysis and treatments are to be applied.
In the biological follow-up of HIV infected patients, plasma (RNA) viral load is the key parameter currently recommended by WHO. In fact it is a good marker of therapeutic adherence, disease progression and treatment efficacy. In addition, according to recent WHO’s recommendations, it will be used as the main therapeutic follow-up parameter rather than CD4 counts.
For HIV plasma viral load analyses, the Virology Department of Centre Pasteur of Cameroon uses two platforms: Generic HIV Viral Load assay (Biocentric, Bandol, France) and Abbott Real-Time HIV-1 assay (Abbott Molecular, Wiesbaden, Germany). The techniques choice is based on the HIV-1 group harbored by the patient. Generic HIV Viral Load assay, used for the quantification of HIV-1 group M is a cheap and open system whose usefulness has been shown in resource limited countries like Cameroon. It has also been shown to be applicable in the context of high viral diversity among HIV-1 group M. Abbott Real-Time HIV-1 assay is broader in terms of range of HIV groups covered , however because of unaffordable cost of analysis, this test is less implemented in resource limited settings. In order to choose an adequate and cheap HIV viral load technique in our laboratory for each sample, HIV types/groups determination is performed routinely using an in-house ELISA assay as previously described . In spite of the systematic application of this discriminatory test, the types/groups of some samples remain indeterminate. The objective of this study was to compare the viral loads of serologically indeterminate HIV samples using two techniques: the Generic HIV Viral Load assay (Biocentric, Bandol France) and the Abbott Real-Time HIV-1 assay (Abbott Molecular, Wiesbaden Germany) with the view of reducing cost in such genetically divergent viral populations.
Description of the study site
The Centre Pasteur of Cameroon (CPC) is one of the reference HIV laboratories in Cameroon. It plays a central role in evaluating HIV tests intended to be used for routine analyses in Cameroon. As part of this role, the CPC determined the utility a cheaper test, the Biocentric method for HIV serological indeterminate samples. All the data were collected in the Virology service from patients who requested HIV viral load analyses at CPC. Two platforms are used for viral load quantification selected according to HIV serotyping results: Biocentric is specifically dedicated for HIV-1 group M quantification while Abbott is used for other HIV-1 variants.
Between March and May 2014, and between August and September 2014, 6355 consecutive samples were registered in the Virology laboratory of Centre Pasteur of Cameroon for viral load analysis. As a routine test, prior to all viral load analysis in our laboratory, HIV-group serotyping was performed on all these samples using an in-house ELISA as previously described . Briefly, this test uses HIV peptides of the V3 loop and gp41/36 regions of HIV-1 groups M, N, O and P as well as HIV-2 peptides. A first test, “two peptides” format, explores the V3 region of HIV-1 groups M (HIV-1 M) and O (HIV-1 O). Negative samples in the “two peptides” assay were re-tested in a second ELISA known as the “ten peptides” format using peptides mapping the gp41/36 region of HIV-1 groups M, O, N and HIV-2/SIVsm on one hand and V3 peptides of HIV-1 groups M, O (subgroup H and consensus), N, P, and HIV-2/SIVsm on the other hand.
Plasma HIV RNA quantification
All HIV-1 M samples identified by serotyping were quantified by the Generic HIV Viral Load assay (Biocentric, Bandol-France), which targets a well-conserved LTR region of HIV-1 . Non-M samples (HIV-1 N, O and P) identified by serotyping were quantified by Abbott Real-Time HIV-1 assay (Abbott molecular, Wiesbaden-Germany), which targets a highly conserved integrase-coding region of the pol gene . For the purpose of this study, HIV group-indeterminate samples (by serotyping) were quantified with both techniques.
Generic HIV Viral Load assay (Biocentric Bandol-France)
RNA was extracted manually from 1 mL of all HIV-1 group M plasma samples using QIAmp viral RNA mini kit (QIAGEN, Courtaboeuf, France) according to the manufacturer instructions. Purified RNA was eluted in 60 μL of molecular grade water. A volume of 10 μL of RNA extracts was therefore used for quantification with the Generic HIV Viral Load assay (Biocentric, Bandol-France) as originally described by Rouet and collaborators in the ANRS HIV quantification working group  and previously reported by our team . The cycling conditions consisted of 50 °C for 10 min and 95 °C for 5 min, followed by 50 cycles of 95 °C for 15 s and 60 °C for 1 min. Amplification and data acquisition were carried out using the ABI Prism 7300 Sequence Detection System (Applied Biosystems) and the detection cut-off value was 60 HIV-1 RNA copies/mL.
Abbott Real-Time HIV-1 assay (Abbott molecular, Wiesbaden-Germany)
This assay was performed on samples that tested HIV-1 Group O or N by serotyping. The test was run according to the manufacturer’s instructions. Briefly, HIV-1 RNA was extracted from 0.6 mL of plasma sample, using the Abbott m2000sp nucleic acid extraction system. 50 μL of purified RNA was subsequently mixed with 50 μL of Master mix and run on the m2000rt Real-Time PCR system . The detection cut-off value was 40 HIV-1 RNA copies/mL.
For the purpose of this study, samples that were serologically HIV type/group-indeterminate were quantified by both the Abbott Real-Time HIV-1 and Generic HIV Viral Load assays. Samples showing detectable viral loads were further characterized using molecular tests.
Molecular tests were performed using an RT-nested PCR targeting the pol gene (Integrase) of HIV-1 group M and O as previously described . RNA extracts originating from samples that were detectable by only one of the quantification techniques (Generic HIV Viral Load and Abbott HIV-1 viral load assays) were further analyzed using an RT-nested PCR targeting the Gp41 region of the Env gene as previously described . This PCR uses primers with a broad specificity and can amplify HIV-1 (M, O, and N) as well as Simian Immunodeficiency Viruses from Chimpanzees (SVIcpz). Primers were designed on the basis of conserved gp41 (gpM–Z) regions. The outer primers gp40F1 and gp41R1 and the inner primers gp46F2 and gp48R2 were used with the PCR conditions previously described .
Sequence and phylogenetic analyses
PCR products were sequenced by the Sanger method using BigDye® Terminator Cycle Sequencing Ready Reaction v3.1 kit (Applied Biosystems). Sequences were aligned by CLUSTALW and phylogenetic trees were inferred using a Kimura two-parameter substitution model and the neighbor-joining method with 1000 bootstrapped data sets implemented with the MEGA6.06 software .
We performed the correlation coefficient of Pearson to observe association between HIV-1 viral loads obtained with the two techniques. Furthermore, the statistical tests of Karle Wallis were implemented to compare the value of correlation with 0. All the statistical analyses were performed using R software version 2.15.
Among the 6355 plasma samples received, HIV-1 group M was identified in 6026 (94.82%) cases, HIV-1 group O in 20 (0.31%), HIV-1 group M + O in 3 (0.05%) and HIV-2 in 3 (0.05%) cases. HIV-group indeterminate samples represented about 4.76% (303/6355) of all samples received.
Abbott real-time HIV-1 and generic HIV viral load assays of HIV group indeterminate samples
Because of the limitation of the volume of some plasma samples, only 231 of the 303 types/groups indeterminate samples were available and could be analyzed by Abbott Real-Time HIV-1 and Generic HIV Viral Load techniques. The results showed that 188 (81.39%) of these samples showed undetectable viral load using both techniques. All the remaining 43 samples were detectable with Abbott Real-Time HIV-1 while 40 of them were also detectable with the Generic HIV Viral Load (Table 1). All detectable samples in both techniques showed high viral load, with a mean of 4.5 log copies/ml (range 2.1–6.5) and 4.5 log copies/ml (range 2–6.4) for Abbott Real-Time and Generic HIV Viral Load, respectively.
The mean viral load difference between the two techniques was 0.03 log10 copies/ml which was not significantly different. Importantly, the difference between both assays did not increase with low or high viral loads. A good correlation was found between the results of both assays (Pearson correlation coefficient r 2 = 0.89; P < 0.001) as shown on Fig. 1.
Molecular characterization of HIV type/group indeterminate samples showed that all forty samples detectable by both techniques were HIV-1 group M while the three discordant samples detected only by the Abbott Real-Time HIV-1 technique were HIV-1 group O. Further phylogenetic analysis of the sequences of the Gp41 region derived from the three HIV-1 group O samples, showed that two of them belonged to HIV-1 group O subgroup H while the last one was a member of the subgroup T (Fig. 2).
Plasma (RNA) viral load is the key parameter currently recommended by WHO for the biological follow-up of HIV infected patients because it is a good marker of therapeutic adherence, disease progression and treatment efficacy. In our laboratory, two platforms are used for viral load quantification selected according to HIV serotyping results: Generic HIV Viral Load (Biocentric) is specifically dedicated for HIV-1 group M quantification and Abbott Real-Time (Abbott) for other HIV-1 variants. Serotyping used for the discrimination of different HIV variants (HIV-1 groups M, N, O and P and HIV-2), is performed using two sets of peptides: the gp41/36 and the V3 specific for each HIV type/group. Data have shown that the gp41/36 peptides have an excellent sensitivity while the V3 peptides have an excellent specificity for the corresponding HIV-1 group, especially the group M peptides . Simon and collaborators demonstrated that the V3 discrimination component of the test clearly identified all the group M samples and there was no V3 reactivity other than that directed to the peptides mapping this group. Serotyping could therefore be systematically performed as complementary test for all HIV-positive samples because this will help for the choice of appropriate treatment option as well as the platform to be used for viral load. Thus, samples classified as HIV-1 group M in our study (6026 out of 6355 samples received) were correctly quantified by Generic HIV Viral Load (Biocentric).
Our serological data confirmed the circulation of HIV-1 (groups M, N and O), HIV-2, as well as HIV-1 M + O dual infections. These data confirm the great genetic diversity of HIV in Cameroon [20,21,22]; with HIV-1 M representing the broad majority of HIV in circulation. It is known that HIV-1 M is the pandemic form of HIV because it is distributed worldwide, while other variants are mostly found in Africa, especially in Cameroon . Our results showed that 0.3% of samples were HIV-1 group O, and this is consistent with the actual epidemiology of HIV-1 group O whose prevalence is estimated at approximately 0.4–1% of all HIV infections in Cameroon [20,21,22]. Among the 231 HIV type/group indeterminate samples analyzed with both Generic HIV Viral Load and Abbott Real-Time assays, as high as 81.39% had undetectable viral load in both techniques. This could be due to the fact that most of patients received in our laboratory for HIV viral load are on Combined Antiretroviral Therapy; and according to the WHO 2013 recommendations which are used in the country, viral load is performed six months after treatment initiation. These undetectable VL results indicate the absence of plasma HIV-1 RNA and reveal expected treatment efficacy for the majority of the patients received. It has been demonstrated that treatment can drop viral load and antibody titers can subsequently decrease leading to indeterminate serological tests  as observed with indeterminate results in serotyping. However, PCR could be used to confirm HIV status in such cases; we previously described seronegative results with positive DNA PCR in children who started treatment early .
We found a good concordance (98.7%) between both techniques, and the majority of samples found to be detectable with Abbott Real-time were also detectable with Generic HIV Viral Load, with a good correlation between viral load values (Pearson correlation coefficient r 2 = 0.89; P < 0.001). This is consistent with results obtained by Rouet and collaborators who described the performance of the ANRS (Agence Nationale de Recherche sur le SIDA) second-generation long terminal repeat-based real-time RT-PCR test, Generic HIV Viral Load, for the quantification of HIV in a context of high HIV genetic diversity . The molecular characterization showed that all 40 samples efficiently detected by both techniques were HIV-1 group M, whereas, the three samples refractory to Generic HIV Viral Load were HIV-1 group O. This result can be explained by the fact that Generic HIV Viral Load has been designed for the quantification of HIV-1 group M and has been shown to inaccurately quantify certain strains of HIV-1 group O . In the phylogenetic analysis, two of HIV-1 group O sequences were closely related and supported by a high bootstrap value (Fig. 2). Further analyses were performed to investigate a potential transmission pair. These two sequences displayed a 97.43% identities and they were obtained from two women from different cities. Also, the patients were received at different date with no epidemiological link. Altogether, these results suggested that there is no transmission pair even if the two sequences are more related to each other than to other HIV-1 group O sequences.
Even though viral load has become the main virological marker for detecting treatment failure, this test is not affordable in many countries especially in the resource limited settings. Because of inadequate laboratory capacity and high cost of equipment, viral loads are performed in the reference laboratory; samples have to be collected in the field and shipped to reference laboratory for testing. In this context, logistic barriers (plasma separation, storage and shipping of specimens) are major challenges in the availability of HIV viral load in peripheral regions. Therefore, there is an urgent need in the development of less costly methods such as viral load pooling, viral load using DBS as well as point of care to monitor patients receiving antiretroviral therapy. These methods could be helpful for the sustainability of the monitoring of patients; however, such strategies should to be validated in each region/country in order to take into consideration the specificity of the designated area.
Nonetheless, our study has some limitations because the presence of non HIV-1 variants in the 188 samples with HIV type/group indeterminate results with serotyping and undetectable viral load on both Abbott Real-time and Generic HIV Viral Load cannot be ruled out. Since these viral load assays could not amplify non HIV-1 variants, nested-PCR for HIV-2 and SIV should be performed on these samples.
In conclusion, our results suggest that a cheaper and open technique such as Generic HIV Viral Load (Biocentric) could be a reliable alternative for HIV viral load follow-up quantification in the settings marked by a high viral diversity, like Cameroon. In such settings, molecular characterization could therefore be performed on samples originating from patients with immuno-virological or immuno-clinical discordances, in order to detect potential non-M HIV variants. Since the price of Abbott Real-Time assy is twice the price of the Generic assay in Cameroon, using this Biocentric platform will save 450,000 US dollars at the end of the year.
Combined Antiretroviral Therapy
Centre Pasteur du Cameroun
Demographic Health Survey
Human Immunodeficiency Virus
Polymerase Chain Reaction
World Health Organization
Gao F, Chen Y, Levy DN, Conway JA, Kepler TB, Hui H. Unselected mutations in the human immunodeficiency virus type 1 genome are mostly nonsynonymous and often deleterious. J Virol. 2004;78:2426–33.
Levy DN, Aldrovandi GM, Kutsch O, Shaw GM. Dynamics of HIV-1 recombination in its natural target cells. Proc Natl Acad Sci U S A. 2004;101:4204–9.
Plantier JC, Leoz M, Dickerson JE, De Oliveira F, Cordonnier F, Lemee V, Damond F, Robertson DL, Simon FA. New human immunodeficiency virus derived from gorillas. Nat Med. 2009;15:871–2.
Robertson DL, Anderson JP, Bradac JA, Carr JK, Foley B, Funkhouser RK, Gao F, Hahn BH, Kalish ML, Kuiken C, Learn GH, Leitner T, McCutchan F, Osmanov S, Peeters M, Pieniazek D, Salminen M, Sharp PM, Wolinsky S, Korber B. HIV-1 nomenclature proposal. Science. 2000;288:55–6.
Ayouba A, Akoua-Koffi C, Calvignac-Spencer S, Esteban A, Locatelli S, Li H, Li Y, Hahn BH, Delaporte E, Leendertz FH, Peeters M. Evidence for continuing cross-species transmission of SIVsmm to humans: characterization of a new HIV-2 lineage in rural cote d'Ivoire. AIDS. 2013;27:2488–91.
ORC-Macro INdlSa. Enquête Démographique et de Santé du Cameroun 2011. 2011.
Plantier JC, Djemai M, Lemee V, Reggiani A, Leoz M, Burc L, Vessiere A, Rousset D, Poveda JD, Henquell C, Gautheret-Dejean A, Barin F. Census and analysis of persistent false-negative results in serological diagnosis of human immunodeficiency virus type 1 group O infections. J Clin Microbiol. 2009;47:2906–11.
Descamps D, Collin G, Letourneur F, Apetrei C, Damond F, Loussert-Ajaka I, Simon F, Saragosti S, Brun-Vezinet F. Susceptibility of human immunodeficiency virus type 1 group O isolates to antiretroviral agents: in vitro phenotypic and genotypic analyses. J Virol. 1997;71:8893–8.
Leoz M, Feyertag F, Kfutwah A, Mauclere P, Lachenal G, Damond F, De Oliveira F, Lemee V, Simon F, Robertson DL, Plantier JC. The two-phase emergence of non pandemic HIV-1 group O in Cameroon. PLoS Pathog. 2015;11:e1005029.
Rouet F, Liegeois F, Mouinga-Ondeme A, Kania D, Viljoen J, Wambua S. Ngo-Giang-Huong N, Menan H, Peeters M, Nerrienet E. Current challenges to viral load testing in the context of emerging genetic diversity of HIV-1. Expert Opin Med Diagn. 2011;5:183–202.
Gueudin M, Leoz M, Lemee V, De Oliveira F, Vessiere A, Kfutwah A, Plantier JCA. New real-time quantitative PCR for diagnosis and monitoring of HIV-1 group O infection. J Clin Microbiol. 2012;50:831–6.
Kfutwah A, Lemee V, Ngono HV, De Oliveira F, Njouom R, Plantier JC. Field evaluation of the Abbott ARCHITECT HIV ag/ab combo immunoassay. J Clin Virol. 2013;58(Suppl 1):e70–5.
Rouet F, Chaix ML, Nerrienet E, Ngo-Giang-Huong N, Plantier JC, Burgard M, Peeters M, Damond F, Ekouevi DK, Msellati P, Ferradini L, Rukobo S, Marechal V, Schvachsa N, Wakrim L, Rafalimanana C, Rakotoambinina B, Viard JP, Seigneurin JM, Rouzioux C. Impact of HIV-1 genetic diversity on plasma HIV-1 RNA quantification: usefulness of the Agence Nationale de Recherches sur le SIDA second-generation long terminal repeat-based real-time reverse transcriptase polymerase chain reaction test. J Acquir Immune Defic Syndr. 2007;45:380–8.
Schutten M, Peters D, Back NK, Beld M, Beuselinck K, Foulongne V, Geretti AM, Pandiani L, Tiemann C, Niesters HG. Multicenter evaluation of the new Abbott RealTime assays for quantitative detection of human immunodeficiency virus type 1 and hepatitis C virus RNA. J Clin Microbiol. 2007;45:1712–7.
Tejiokem MC, Faye A, Penda IC, Guemkam G, Ateba Ndongo F, Chewa G, Rekacewicz C, Rousset D, Kfutwah A, Boisier P, Warszawski J. Feasibility of early infant diagnosis of HIV in resource-limited settings: the ANRS 12140-PEDIACAM study in Cameroon. PLoS One. 2011;6:e21840.
Heyndrickx l, Janssens w, Gürtler l, Zekeng l, Loussert-ajaka I, Vereecken K, Willems B, Coppens S, Ndumbe P, Fransen K, Saman E, Alary M, Groen1 gvd. Differential diagnosis of HIV type 1 group O and MInfection by polymerase chain reaction and Pstl restriction analysis of the pol gene fragment. AIDS Res Hum Retrovir. 14:973–7.
Yang C, Dash BC, Simon F, van der Groen G, Pieniazek D, Gao F, Hahn BH, Lal RB. Detection of diverse variants of human immunodeficiency virus-1 groups M, N, and O and simian immunodeficiency viruses from chimpanzees by using generic pol and env primer pairs. J Infect Dis 2000;181:1791–1795.
Tamura K, Stecher G, Peterson D, Filipski A, Kumar S. MEGA6: Molecular Evolutionary Genetics Analysis version 6.0. Mol Biol Evol. 2013;30:2725–9.
Simon F, Souquiere S, Damond F, Kfutwah A, Makuwa M, Leroy E, Rouquet P, Berthier JL, Rigoulet J, Lecu A, Telfer PT, Pandrea I, Plantier JC, Barre-Sinoussi F, Roques P, Muller-Trutwin MC, Apetrei C. Synthetic peptide strategy for the detection of and discrimination among highly divergent primate lentiviruses. AIDS Res Hum Retrovir. 2001;17:937–52.
De Oliveira F, Mourez T, Vessiere A, Ngoupo PA, Alessandri-Gradt E, Simon F, Rousset D, Plantier JC. Multiple HIV-1/M + HIV-1/O dual infections and new HIV-1/MO inter-group recombinant forms detected in Cameroon. Retrovirology. 2017;14:1.
Villabona-Arenas CJ, Domyeum J, Mouacha F, Butel C, Delaporte E, Peeters M, Mpoudi-Ngole E, Aghokeng AF. HIV-1 group O infection in Cameroon from 2006 to 2013: prevalence, genetic diversity, evolution and public health challenges. Infect Genet Evol. 2015;
Vessiere A, Rousset D, Kfutwah A, Leoz M, Depatureaux A, Simon F, Plantier JC. Diagnosis and monitoring of HIV-1 group O-infected patients in Cameroun. J Acquir Immune Defic Syndr. 2010;53:107–10.
Peeters M, Toure-Kane C, Nkengasong JN. Genetic diversity of HIV in Africa: impact on diagnosis, treatment, vaccine development and trials. AIDS. 2003;17:2547–60.
Bannister WP, Ruiz L, Loveday C, Vella S, Zilmer K, Kjaer J, Knysz B, Phillips AN, Mocroft A. HIV-1 subtypes and response to combination antiretroviral therapy in Europe. Antivir Ther. 2006;11:707–15.
Kfutwah AK, Tejiokem MC, Ateba FN, Ndongo JA, Penda IC, Ngoupo PA, Tchendjou P, Chewa G, Boisier P, Rouzioux C, Warszawski J, Faye A. Seronegativation in early treated HIV-infected infants: frequency and potential implications on care and follow-up in a resource-limited country. J Acquir Immune Defic Syndr. 2011;58:e43–6.
We would like to thank all the patients for the data provided. All the data presented here were anonymously analysed and reported. We also like to thank the Centre Pasteur of Cameroon for its financial support.
This work was supported by the Centre Pasteur of Cameroon. CPC was not involved in the study design, collection of samples, and interpretation of data or preparation of the manuscript.
Availability of data and materials
The nucleotide sequences of HIV-1 group O described from the discordant samples are in process for submission to GenBank.
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
About this article
Cite this article
Ngo-Malabo, E., Ngoupo T., P.A., Zekeng, M. et al. A cheap and open HIV viral load technique applicable in routine analysis in a resource limited setting with a wide HIV genetic diversity. Virol J 14, 224 (2017). https://doi.org/10.1186/s12985-017-0893-3
- HIV viral load quantification
- Serotyping- group indeterminate
- Resource limited setting
- Generic HIV viral load
- Abbott real-time HIV-1