- Open Access
Development of an improved microneutralization assay for respiratory syncytial virus by automated plaque counting using imaging analysis
© Zielinska et al; licensee BioMed Central Ltd. 2005
Received: 08 June 2005
Accepted: 09 November 2005
Published: 09 November 2005
Respiratory syncytial virus (RSV) is the major cause of lower respiratory tract infection in infants and young children. Although several experimental RSV vaccines are under investigation, immuno therapy is the only treatment currently available. In assessing the immunogenicity of various vaccine formulations, a plaque reduction neutralization assay for the evaluation of RSV neutralizing antibody has been widely used. The method produces reliable results, but it is tedious and labor intensive as it relies on manual counting by laboratory personnel. To facilitate evaluation of phase II and phase III vaccine clinical trials, a more rapid, reliable and efficient neutralization assay is needed.
An improved microneutralization assay for quantifying RSV neutralizing antibodies was developed using an ImmunoSpot® Series I Analyzer (Cellular Technology Ltd., Cleveland, OH) for automated plaque counting. The method is an improvement of the established classical microneutralization assay in which immunostained plaques on transparent tissue culture plates are counted manually under a dissecting microscope. Image analyzer technology allows for fully automated counting of plaques distributed throughout an entire well. Adjustments, such as the use of opaque tissue culture plates and the TMB substrate, True Blue™ (KPL, Gaithersburg, MD), were required to adapt the assay for optimal detection of plaques by the image analyzer. The suitability and the accuracy of the method for counting RSV plaques were determined by comparative testing of a reference serum and two control sera by manual and automated counting methods. The results showed that the two methods were highly correlated (R = 0.9580) and the titers generated by them were within two-fold.
Our results demonstrate that the semi-automated assay is rapid and reliable. It provides results within two fold to the classical plaque microneutralization assay and is readily applied to the evaluation of neutralizing antibody titers in sera obtained from epidemiology or vaccine clinical trials.
Respiratory syncytial virus (RSV) is the most common form of lower respiratory viral infection affecting infants, the elderly, and immunocompromised individuals . In severe cases, it may cause complications ranging from pneumonia and bronchiolitis to death . While the most severe outcomes arise in patients with weakened or underdeveloped immune systems, RSV is also gaining notoriety as an important player in annual respiratory disease epidemics among healthy adults . Consequently, there is an obvious unmet need for an efficacious vaccine. The development of a vaccine will require intensive evaluation of the immune response, which can be expedited by utilizing automation.
The neutralization assay is one of the most trusted and widely used methods employed for the detection of virus-specific neutralizing antibodies . The power of the neutralization assay lies in its ability to detect biologically active antibodies. While there are many methods that provide information about different aspects of the immune response (e.g. cellular immunity, genetic markers, etc.) the neutralization assay remains a proven indicator of serological immunity for many viruses. In practice, however, the plaque neutralization assay is a laborious and time-consuming procedure, making it less suitable for testing the large numbers of samples that are obtained in clinical trials. Here, we demonstrate the utility of a method that automates the most laborious and subjective part of the serum neutralization assay – the determination of plaque number. Our results show good agreement between the visual and automated high throughput counting methods for determining RSV serum neutralization antibody titers.
These results were presented previously at the VI International Symposium on Respiratory Viral Infections, March 18–21, 2004, Fort Myers, Florida 
Summary of reference serum titers in logarithm base 4 scale.
The data presented here demonstrate agreement and equivalence between traditional manual and automated plaque counting methods for detection of RSV neutralizing antibody titers. The 180 tests performed in the presence and absence of complement demonstrate that a wide range of titers is accurately detectable by both assays. The automated counting method does not increase the overall variability of the assay; rather variability was observed to be slightly lower with the automated method. Furthermore, we established that the two methods generated titers within 2-fold of each other. A clinically significant change of titer for an RSV patient is indicated by a 4-fold increase in titer, indicative of seroconversion . Therefore, equivalence within 2-fold provides an acceptable level of confidence for the automated counting method.
The strength of this method is that it combines established plaque neutralization procedures with the technology of computerized image scanning and analysis. It has the advantage of providing more efficient and objective results while automating the most laborious and subjective aspect of the assay – plaque counting. Results can be stored as images and plaque counts indefinitely. This allows for better tracking of raw data, as is now mandated by federal and international regulatory bodies. Another important strength of this counting system is that it is capable of detecting and differentiating plaques of different morphology and thus can be used to assay many different viruses. In fact, we have already verified the capability of this system to read plaques created by mumps, influenza and other viruses (data not shown) whether in the context of determining viral potency or performing neutralization assays.
With the technological advances now available, the speed of plate scanning can be reduced further from 15 minutes/plate to approximately 2 minutes/plate. Robotic automation of plate loading can be introduced for further efficiency. The utilization of the TMB substrate staining also facilitates conservation of primary and secondary antibody stocks, which is an important factor when using viruses for which specific antibodies are not readily or commercially available. The objectivity and efficiency provided by this method of plaque counting facilitates the determination of RSV neutralizing antibody titers and can be readily applied to human epidemiology and vaccine clinical studies.
In this report, we describe an RSV microneutralization assay that relies on automated plaque counting and provides a more rapid and less laborious method for detecting neutralizing antibodies to RSV. It provides equivalent results to the classical plaque neutralization assay and can be used in epidemiology and vaccine clinical studies.
Materials and methods
Human sera provided by Intergen Bio-Diagnostics (Purchase, NY) and Bioreclamation Inc. (Hicksville, NY), were tested for anti-RSV antibody titers. Sera were selected and pooled into 3 groups according to titer. The three groups consisted of a lyophilized reference serum, prepared under the auspices of NIAID and two control sera of high and low titer. An historical in-house control standard, C587645, was also tested. The reference serum made up the majority of tests (136 tests), whereas the control sera were each tested approximately 14 times. These sources of human sera comprised the specimens evaluated in this study and were collectively tested 180 times by both methods, in the presence and absence of complement.
Virus and cells
The A2 strain of RSV was used as the challenge virus in all tests. Vero cells (ATCC Cat #CCL 34, ATCC, Rockville, MD) were cultured in EMEM with L-glutamine, 10% FBS, 1% of antibiotic/antimycotic, and non-essential amino acids. Cells were cultured on 96-well white opaque tissue culture plates (BD Falcon, Bedford, MA) for automated counting and on transparent 96-well plates (Corning-Costar, Corning, NY) for manual plaque counting 1–3 days prior to infection.
Determination of RSV antibody titers
Serum samples were heat-inactivated at 56°C for 30 minutes. Four-fold serial dilutions from 1:10 to 1:10,240 were prepared in virus diluent (EMEM with L-glutamine containing 2% FBS, 2.5% HEPES (1 M) and 1% antibiotic/antimycotic, 100×). All sera were tested in the presence and absence of 10% guinea pig complement (Cambrex/BioWhittaker, Walkersville, MD) which was added to the virus diluent prior to the addition of challenge virus. Serially diluted serum was challenged with an equal volume of the RSV-A2 strain, previously titered to give 50–100 pfu per 50 μl of inoculum. The serum/virus mixtures were incubated at 37°C, 5% CO2 for 1 h.
Vero cell monolayers, prepared in 96 well plates, were infected with 50 μl/well (in duplicate) of the serum/virus mixture. Plates were centrifuged at 1 h at 2000 rpm (700 g), followed by 30 min of rocking at room temperature. Supernatants were decanted. Plates were blotted and overlaid with 0.75% methyl cellulose (4,000 cP at 2% aqueous), prepared in MEM with 2% FBS, warmed to 37°C and inoculated at 100 μl/well. Plates were incubated at 37°C, 5% CO2 for 3 days to allow for plaque formation.
Conventional staining and plaque determination for RSV neutralization
Cells infected on transparent plates were fixed with a 50%:50% methanol:ethanol mixture at room temperature for 10 min. Plates were washed with DPBS after fixing and between staining steps. Plates were incubated for 1 h at 37°C, 5% CO2 with 50 μl/well of monoclonal antibody specific for RSV-A2 F-protein (Wyeth K6-5-1) diluted to 1:1,000 in Blotto (5% non-fat milk in PBS). Peroxidase labeled secondary goat anti-mouse IgG antibody (KPL, Gaithersburg, MD) diluted 1:100 in Blotto, was incubated at 50 μl/well for 1 h at room temperature. Plaques were developed using 100 μl/well 3,3'diaminobenzidine HRP substrate (0.5 mg/ml DAB, 0.01% H2O2) prepared in DPBS and incubated at room temperature for 5 – 10 minutes. Plates were washed with tap water to stop the reaction.
Plaques were counted manually by inverting the transparent plate under a dissecting microscope. The field of the well was separated into quadrants for ease of counting. Overlapping plaques were deemed individual when lobes were apparent.
TMB staining and plaque determination for RSV neutralization
Cells infected on opaque white tissue culture plates were fixed and washed as described above. Plates were incubated for 1 h at 37°C 5% CO2 with 50 μl/well of monoclonal antibody specific for RSV-A2 F-protein (Wyeth, K6-5-1) diluted to 1:10,000 in blotto. Peroxidase labeled secondary goat anti-mouse IgG antibody (KPL, Gaithersburg, MD) diluted 1:3,000 in Blotto was added at 50 μl/well and incubated for 1 hour at room temperature. Plaques were developed using 50 μl/well of a ready to use TMB precipitate HRP substrate, True Blue™ (KPL, Gaithersburg, MD). Higher dilutions of primary and secondary antibody were used due to the increased sensitivity of the peroxidase TMB substrate. Plates were washed thoroughly with tap water to stop the reaction and dried inverted in order to minimize bleaching.
Calculation of antibody titer
Titer was calculated from the average of duplicate sample wells by extrapolating the inverse dilution of serum that produced a 60% reduction of virus according to the following formula:
X = (a-b)(e-c)/(c-d) + a
where, a = log10 of dilution above the 60% reduction point, b = log10 of dilution below the 60% reduction point, c = average plaque count above the 60% reduction point (corresponds with a), d = average plaque count below the 60% reduction point (corresponds with b) and e = value of 60% reduction of average virus control count.
All titers were reported in logarithm base 4 scale in order to visually represent a difference of one dilution (of a 4-fold dilution series) as 1 log unit. Different statistical analyses were performed to assess the agreement of titers generated by two methods. In one analysis we graphically inspected the spread of the paired titers about the 45° line and computed Pearson's correlation coefficient. In another analysis, we determined the level of equivalence between the two assays, by constructing a difference-means plot .
- Collins PL, Chanock RM, Murphy BR: Respiratory syncytial virus. In In Fields Virology. Volume 1. 4th edition. Edited by: Knipe DM, Howley PM. Philadelphia: Lippincott Williams and Wilkins; 2001:1443-1485.Google Scholar
- Han LL, Alexander JP, Anderson LJ: Respiratory syncytial virus pneumonia among the elderly: an assessment of disease burden. J Infect Dis 1999, 179: 25-30. 10.1086/314567View ArticlePubMedGoogle Scholar
- Zambon M, Stockton JD, Clewley JP, Fleming DM: Contribution of influenza and respiratory syncytial virus to community cases of influenza-like illness: an observational study. Lancet 2001, 358: 1410-1416. 10.1016/S0140-6736(01)06528-XView ArticlePubMedGoogle Scholar
- Casals J: Immunological techniques for animal viruses. In Methods in Virology. Volume III. Edited by: Maramorosch K, Koprowski H. Academic Press, New York; 1967:113-194.Google Scholar
- Zielinska E, Liu D, Wu H, Quiroz J, Rappaport R, Yang DP: A novel high throughput microneutralization assay for respiratory syncytial virus using an ImmuneSpot Analyzer. [Abstract]. VI International Symposium on Respiratory Vial Infections, March 18–21, 2004, Fort Myers, FloridaGoogle Scholar
- Piedra PA, Jewell AM, Cron SG, Atmar RL, Glezen WP: Correlates of immunity to respiratory syncytial virus (RSV) associated-hospitalization: establishment of minimum protective threshold levels of serum neutralizing antibodies. Vaccine 2003, 21: 3479-3482. 10.1016/S0264-410X(03)00355-4View ArticlePubMedGoogle Scholar
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet 307-310. 1986 8 FebGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.