- Short report
- Open Access
Evaluation of four commercial real-time RT-PCR kits for the detection of dengue viruses in clinical samples
Virology Journal volume 11, Article number: 164 (2014)
Dengue is the most frequent arthropod-borne viral disease worldwide. Because dengue manifestations are similar to those of many other febrile syndromes, the availability of dengue-specific laboratory tests is useful for the differential diagnosis. Timely and accurate diagnosis of dengue virus (DENV) infection is important for appropriate management of complications, pathophysiological studies, epidemiological investigations and optimization of vector-control measures. Several “in-house” reverse transcriptase-polymerase chain reaction (RT-PCR) methods have been developed to detect, type and/or quantify DENV. Standardized dengue RT-PCR kits with internal controls have been recently introduced, but need clinical evaluation. We assessed the performances of 4 commercial DENV real-time RT-PCR kits.
The 4 kits were evaluated using a panel of 162 samples positive with an existing in-place hemi-nested RT-PCR used for routine DENV-infection diagnosis in patients with acute-febrile disease. The panel included 46 DENV-1, 37 DENV-2, 33 DENV-3, and 46 DENV-4. Also, 70 negative serum specimens were used to determine specificity. Geno-Sen’s Dengue 1–4 Real-Time RT-PCR kit was the only assay to provide quantification using standards, but lacked sensitivity for DENV-4 detection. The SimplexaTM Dengue RT-PCR assay, with 151 (93.2% [95% confidence interval, 89.3–97.1]) positive samples, had significantly higher sensitivity than the other 3 kits; in a complementary evaluation of 111 consecutive patients’ samples, its performance and genotyping agreed with the hemi-nested gold-standard assay.
The SimplexaTM Dengue RT-PCR’s good performance to detect and genotype DENV1–4 requires further evaluation in multicenter and prospective studies, particularly in settings of clinical diagnosis during dengue outbreaks.
Dengue is caused by 4 related viruses, (DENV)-1, -2, -3 and -4, and is the most common and widespread arthropod-borne viral disease in the world. This self-limited infection can be asymptomatic or cause a clinical spectrum ranging from mild fever (DF) to severe, life-threatening illness, dengue hemorrhagic fever (DHF) and dengue shock syndrome (DSS). Other severe manifestations, including hepatitis, myocarditis or encephalitis, may occur independently of plasma leakage . Because dengue symptoms are similar to those of many other undifferentiated febrile syndromes, new commercial tests that can be used for case management have to be clinically evaluated. Laboratory-based surveillance is also of importance for early warning of dengue outbreaks and optimization of vector-control measures .
Current diagnostic methods have limitations. Enzyme-linked immunosorbent assays to detect immunoglobulin M are used by most laboratories, but that response is absent early during the disease course and may remain undetectable in secondary dengue. DENV-NS1 antigen can be detected during the first few days of fever. However, recent evaluations of commercial NS1 assays have had limited sensitivity [3, 4]. Virus isolation from acute-phase sera is useless for patient management, because several days are needed to obtain final results. Molecular methods yield same- or next-day detection of DENV in acute-phase serum or plasma. The 2-step hemi-nested reverse transcriptase-polymerase chain reaction (RT-PCR) protocol, originally reported by Lanciotti et al.  and later modified to a single-step multiplex RT-PCR , has proved to be highly sensitive , and is used worldwide. Conventional RT-PCR is being progressively replaced by real-time RT-PCR, which limits the risks of carryover contamination and is suitable for DENV-genome detection. Another advantage of real-time RT-PCR is that it yields reliable quantification, making it a useful tool for pathophysiological studies. Almost all of the reported RT-PCR methods are “in-house” procedures [8–29]. International external quality studies have highlighted sensitivity and specificity heterogeneities, even inter-laboratory using the same method .
Levi et al.  evaluated a commercial real-time RT-PCR kit (RealArt; artus/Qiagen, Germany) and found high agreement with an in-house multiplex RT-PCR, but the test was limited to DENV-3, and occasionally DENV-2, samples. More recently, the US Food and Drug Administration approved a Centers for Disease Control DENV-1–4 real-time RT-PCR assay , but it has been shown to be less sensitive than a laboratory-developed assay, particularly for DENV-1 . Herein, we evaluated the performances of 4 recently developed commercial DENV real-time RT-PCR kits.
The study was conducted in the Virology Laboratory, where biological samples from patients with acute-febrile syndromes are processed as follows: a 5 ml venous blood sample, collected on the day of admission, is centrifuged, and plasma or serum supernatant is divided into 2 aliquots, one stored at -4°C for daily RT-PCR and the other frozen at -80°C until use. We analyzed 162 laboratory-confirmed stored samples collected during the outbreaks that occurred the last decade: 46 DENV-1, 37 DENV-2, 33 DENV-3, and 46 DENV-4. A panel of 70 negative specimens was used to determine specificity. In addition, we prospectively tested 111 samples from patients consecutively seen at the emergency unit for acute-febrile syndromes at the peak of the 2010–2011 epidemic.
Because the study was non-interventional, e.g. with no additional samplings or specific procedures for subjects, no consent form from the patient was needed, as stated by the French Public Health code. The consecutive samplings were part of routine diagnosis. The dengue sample collection has been declared under the number DC-2009-979 to the French ministry of health and the “Comité de protection des personnes Bordeaux sud-ouest/Outre-mer”, according to the Bioethics laws (décret n°2007-1220 du 10 août 2007 relatif aux activités de prélèvement et conservation à des fins scientifiques de collections d’échantillons biologiques). The dengue collection is stored at the “Centre de Ressources Biologiques de la Martinique” (CeRBiM). The CeRBiM database has been agreed by the CNIL (“Commission nationale de l’informatique et des libertés”).
RNA was extracted from 200 μL of stored aliquots using NuCliSens EasyMag (bioMérieux, Craponne, France). All samples were screened by hemi-nested RT-PCR with the dengue virus consensus primers D1 and D2 and the serotype-specific primers TS1, TS2, TS3, and TS4 . Lanciotti’s protocol was adapted using the SuperScript One-Step RT-PCR System with Platinum Taq DNA Polymerase (Invitrogen, Life Technologies, France) . Briefly, the first-round reaction was performed with primers D1 and D2, 1 μl of SuperScript II RT/Platinum Taq mix (Invitrogen), and 5 μl of the RNA template. The second amplification was used forward primer D1 and downstream primers TS1, TS2, TS3, and TS4, with the TaqPCR Master Mix kit (Qiagen). Both the RT-PCR and nested-PCR products were analyzed by gel electrophoresis on a 2% agarose gel with ethidium bromide for ultraviolet visualization. The expected size of the primary PCR amplicons was 511 bp. Nested-PCR bands of 482, 119, 290, and 389 bp were considered specific for DENV-1, DENV-2, DENV-3, and DENV-4, respectively. The sensitivity of the assay has been previously evaluated to 1 PFU/ml with dengue virus titrated controls, and 1–10 molecules/reaction with dengue virus RNA standards obtained by cloning the capsid-prM region into pCR4-TOPO vector (data not shown). Finally, the procedure was validated through participation in a multicenter quality-control study conducted by the French National Reference Center for Arboviruses (Institut de Recherche Biologique des Armées, France).
Four commercial kits were evaluated versus Lanciotti’s gold-standard protocol. All the commercial kits were fully furnished, and were used according to the manufacturer’s recommendations. The characteristics of the 4 kits tested are summarized in Table 1.
SimplexaTMdengue RT-PCR assay (Focus Diagnostics, Cypress, CA)
The assay is based on bi-functional fluorescent-probe primers and reverse primers amplifying DENV-1 NS5, DENV-2 NS3, DENV-3 NS5, and DENV-4 capsid genes. The Simplexa dengue 1-and-4–primer mix contains a DENV-1–fluorescent 6-carboxyfluorescein (FAM) probe and a DENV-4–CFR610 probe. The Simplexa dengue 2-and-3–primer mix contains a DENV-3–FAM probe and a DENV-2–CFR610 probe. Each primer mix includes a Q670-labeled probe to detect the RNA internal control (IC) used to monitor the extraction process and RT-PCR inhibition. A positive control is provided in the kit.
RealStar dengue RT-PCR kit 1.0 (Altona Diagnostics, Hamburg, Germany)
This assay is based on the amplification of DENV-specific target sequences and target-specific probes labeled with FAM reporter and quencher dyes. The probe specific to the IC target is labeled with the fluorophore JOE. The IC is added to the specimen/lysis buffer mixture. Primers and probes, based on 2010 sequence alignments of all 4 DENV, detect but do not genotype them.
Dengue virus general type real-time RT-PCR kit LiferiverTM(Shanghai ZJ Bio-Tech Co, China)
This kit contains a super mix for the specific amplification of DENV-1–4 RNA. DENV amplicons are detected using a probe carrying a FAM fluorophore and BHQ1 quencher. The kit contains a system to identify possible RT-PCR inhibition by measuring the VIC/JOE fluorescence of the IC. The supplied external positive control, defined as 1 × 107 copies/mL, enables determination of the gene load.
Geno-Sen’s dengue 1–4 real-time RT-PCR kit (Genome Diagnostics Pvt, New Delhi, India)
The specific master mix contains reagents and enzymes for the specific amplification of DENV-1–4 and the direct detection of the specific amplicon using the TaqMan principle and FAM channel. An IC is provided to detect inhibitors in the extract. Five external positive standards, range 104–108 copies/mL, are supplied for determination of the gene load. The quantitation standards provided in the kit are treated in the same way as extracted samples. This method was initially designed for use in the Rotor Gene Thermocycler and we adapted it to an ABI 7500 apparatus.
In accordance with conventional clinical pratice, amplification and detection were performed in duplicate, using an ABI Prism® 7500 (Applied Biosystems, France) for LifeRiver, Real Star and GenoSen’s kits. For Simplexa, we used the 3 M integrated cycler provided by Focus. Negative and extraction controls were added at each step to avoid contamination and eliminate inhibition.
The Liferiver kit had poor sensitivity on the initial panel of 40 positive samples, detecting only 28 of them, and was not further evaluated. The Geno-Sen’s, RealStar and Simplexa kits were tested with a panel of 162 sera, and were positive for 138 (85.2%), 135 (83.3%), and 151 (93.2%), respectively (Table 2). Simplexa sensitivity was significantly higher than those of Geno-Sen’s (chi-square, P = 0.02) and RealStar kits (P = 0.006). Simplexa sensitivities were homogeneous (range 90.9%–95.6%) from one serotype to another. For all positive samples, serotype identification was concordant with Lanciotti’s hemi-nested round. RealStar achieved adequate sensitivity for DENV-3 and DENV-2 samples, but failed to detect one-fifth of DENV-1 and DENV-4 samples.
Geno-Sen’s was the only kit to provide quantification using standards, with median virus loads of 5 × 107, 1 × 107, 2 × 107 and 1 × 105 copies/mL, for DENV-1, -2, -3 and -4, respectively. We cannot exclude that the sampling was biased towards higher loads, since it concerned symptomatic patients. Viral loads were markedly lower for DENV-4 samples. Notably, low DENV-4 loads in infections were previously reported [17, 24, 35]. Using an in-house quantification method during a DENV-1–DENV-4 co-epidemic, we confirmed finding lower DENV-4 than DENV-1 loads (data not shown). However, we cannot exclude faulty DENV-4 amplification with Geno-Sen’s kit, and its 71% detection rate is insufficient for clinical diagnoses.
The analytical sensitivity of the Simplexa kit versus Lanciotti’s procedure was also evaluated using dilutions of DENV-1–4 strains. For all dilutions tested, Simplexa assay sensitivity was similar to that of the hemi-nested RT-PCR, with differences <0.5 log10. The Simplexa kit was then tested prospectively on 111 successive patients’ samples. All samples positive with the hemi-nested RT-PCR (3 DENV-1, 16 DENV-2, 6 DENV-4) were also positive with the Simplexa kit, with cycle threshold below 31 cycles. One negative sample with a cycle threshold of 38 was negative with both methods after re-extraction.
Specificity was assessed using 70 samples negative with Lanciotti’s RT-PCR. All were negative with Geno-Sen’s and Simplexa kits, but 1 was weekly positive with RealStar. Since no other flavivirus is known to circulate at significant level in inhabitants of our Caribbean island , we did not test for cross-reactivity with clinical specimens of other member of the genus. West Nile virus lineage I and II controls (AccuType, SeraCare, Life Sciences) were checked negative with the Dengue Simplexa RT-PCR kit. The recent outbreak of chikungunya did allow us to verify the absence of cross-reactivity of the Dengue Simplexa assay with this other arbovirus (data not shown). This is of particular interest for countries threatened by its recent emergence in the Americas. Another limitation of our evaluation is the use of Caribbean DENV, which are only representative of the Americas strains and not of DENV circulating in other part of the world.
In summary, RT-PCR–based methods are widely applied to obtain clinical diagnoses of infectious diseases, and several in-house protocols have been evaluated for DENV-RNA detection. Different reagent suppliers, reaction conditions and technical skills may influence protocol reproducibility from one laboratory to another. External quality-assurance studies showed that differences in the operational procedures led to discrepant results between reference laboratories using the same in-house methods . Commercial kits developed under good manufacturing practices may offer improved reliability. They correspond better to quality-insurance rules for clinical diagnoses and include IC. Manufactured DENV RT-PCR kits may provide broader access, and be less time-consuming and more cost-effective for clinical diagnoses. Our preliminary results showed good agreement between the SimplexaTM Dengue RT-PCR assay and Lanciotti’s reference hemi-nested RT-PCR, currently used in our laboratory. That kit achieved >90% detection of each serotype, with clinical samples obtained between days 1 and 7 after fever onset, and concordance for genotyping. We think that the combination of automated extraction and the SimplexaTM RT-PCR kit should be evaluated in multicenter and prospective studies in the context of clinical management during dengue outbreaks.
Ethical Approval was not needed for this study, in line with French regulations.
Reverse transcription-polymerase chain reaction
Thomas L, Moravie V, Besnier F, Valentino R, Kaidomar S, Coquet LV, Najioullah F, Lengellé F, Césaire R, Cabié A, Working Group on Dengue: Clinical presentation of dengue among patients admitted to the adult emergency department of a tertiary care hospital in Martinique: implications for triage, management, and reporting. Ann Emerg Med 2012, 59: 42-50. 10.1016/j.annemergmed.2011.08.010
Peeling RW, Artsob H, Pelegrino JL, Buchy P, Cardosa MJ, Devi S, Enria DA, Farrar J, Gubler DJ, Guzman MG, Halstead SB, Hunsperger E, Kliks S, Margolis HS, Nathanson CM, Nguyen VC, Rizzo N, Vázquez S, Yoksan S: Evaluation of diagnostic tests: dengue. Nat Rev Microbiol 2010, 8: S30-S37. 10.1038/nrmicro2459
Guzman MG, Jaenisch T, Gaczkowski R, Ty Hang VT, Sekaran SD, Kroeger A, Vazquez S, Ruiz D, Martinez E, Mercado JC, Balmaseda A, Harris E, Dimano E, Leano PSA, Yoksan S, Villegas E, Benduzu H, Villalobos I, Farrar J, Simmons CP: Multi-country evaluation of the sensitivity and specificity of two commercially-available NS1 ELISA assays for dengue diagnosis. PLoS Negl Trop Dis 2010.,4(8): doi:10.1371/journal.pntd.0000811
Najioullah F, Combet E, Paturel L, Martial J, Koulmann L, Thomas L, Hatchuel Y, Cabié A, Cesaire R: Prospective evaluation of nonstructural 1 enzyme-linked immunosorbent assay and rapid immunochromatographic tests to detect dengue virus in patients with acute febrile illness. Diagn Microbiol Infect Dis 2011, 69: 172-178. 10.1016/j.diagmicrobio.2010.09.021
Lanciotti RS, Calisher CH, Gubler DJ, Chang GJ, Vorndam AV: Rapid detection and typing of dengue viruses from clinical samples by using reverse transcriptase-polymerase chain reaction. J Clin Microbiol 1992, 30: 545-551.
Harris E, Roberts TG, Smith L, Selle J, Kramer LD, Valle S, Sandoval E, Balmaseda A: Typing of dengue viruses in clinical specimens and mosquitoes by single-tube multiplex reverse transcriptase PCR. J Clin Microbiol 1998, 36: 2634-2639.
Raengsakulrach B, Nisalak A, Maneekarn N, Yenchitsomanus P-T, Limsomwong C, Jairungsri A, Thirawuth V, Green S, Kalayanarooj S, Suntayakorn S, Sittisombut N, Malasit P, Vaughn D: Comparison of four reverse transcription-polymerase chain reaction procedures for the detection of dengue virus in clinical specimens. J Virol Methods 2002, 105: 219-232. 10.1016/S0166-0934(02)00104-0
Callahan JD, Wu SJ, Dion-Schultz A, Mangold BE, Peruski LF, Watts DM, Porter KR, Murphy GR, Suharyono W, King CC, Hayes CG, Temenak JJ: Development and evaluation of serotype- and group-specific fluorogenic reverse transcriptase PCR (TaqMan) assays for dengue virus. J Clin Microbiol 2001, 39: 4119-4124. 10.1128/JCM.39.11.4119-4124.2001
Chien L-J, Liao T-L, Shu P-Y, Huang J-H, Gubler DJ, Chang G-JJ: Development of real-time reverse transcriptase PCR assays to detect and serotype dengue viruses. J Clin Microbiol 2006, 44: 1295-1304. 10.1128/JCM.44.4.1295-1304.2006
Chutinimitkul S, Payungporn S, Theamboonlers A, Poovorawan Y: Dengue typing assay based on real-time PCR using SYBR Green I. J Virol Methods 2005, 129: 8-15. 10.1016/j.jviromet.2005.05.006
Conceição TM, Da Poian AT, Sorgine MHF: A real-time PCR procedure for detection of dengue virus serotypes 1, 2, and 3, and their quantitation in clinical and laboratory samples. J Virol Methods 2010, 163: 1-9. 10.1016/j.jviromet.2009.10.001
Dumoulin A, Marti H, Panning M, Hatz C, Hirsch HH: Pan-dengue virus detection by PCR for travelers returning from the tropics. J Clin Microbiol 2008, 46: 3104-3106. 10.1128/JCM.01294-08
Gomes-Ruiz AC, Nascimento RT, de Paula SO, da Fonseca BA L: SYBR green and TaqMan real-time PCR assays are equivalent for the diagnosis of dengue virus type 3 infections. J Med Virol 2006, 78: 760-763. 10.1002/jmv.20620
Gurukumar KR, Priyadarshini D, Patil JA, Bhagat A, Singh A, Shah PS, Cecilia D: Development of real time PCR for detection and quantitation of Dengue Viruses. Virol J 2009, 6: 10. 10.1186/1743-422X-6-10
Hue KDT, Tuan TV, Thi HTN, Bich CTN, Anh HHL, Wills BA, Simmons CP: Validation of an internally controlled one-step real-time multiplex RT-PCR assay for the detection and quantitation of dengue virus RNA in plasma. J Virol Methods 2011, 177: 168-173. 10.1016/j.jviromet.2011.08.002
Ito M, Takasaki T, Yamada K-I, Nerome R, Tajima S, Kurane I: Development and evaluation of fluorogenic TaqMan reverse transcriptase PCR assays for detection of dengue virus types 1 to 4. J Clin Microbiol 2004, 42: 5935-5937. 10.1128/JCM.42.12.5935-5937.2004
Johnson BW, Russell BJ, Lanciotti RS: Serotype-specific detection of dengue viruses in a fourplex real-time reverse transcriptase PCR assay. J Clin Microbiol 2005, 43: 4977-4983. 10.1128/JCM.43.10.4977-4983.2005
Kong YY, Thay CH, Tin TC, Devi S: Rapid detection, serotyping and quantitation of dengue viruses by TaqMan real-time one-step RT-PCR. J Virol Methods 2006, 138: 123-130. 10.1016/j.jviromet.2006.08.003
Lai Y-L, Chung Y-K, Tan H-C, Yap H-F, Yap G, Ooi E-E, Ng L-C: Cost-effective real-time reverse transcriptase PCR (RT-PCR) to screen for Dengue virus followed by rapid single-tube multiplex RT-PCR for serotyping of the virus. J Clin Microbiol 2007, 45: 935-941. 10.1128/JCM.01258-06
Laue T, Emmerich P, Schmitz H: Detection of dengue virus RNA in patients after primary or secondary dengue infection by using the TaqMan automated amplification system. J Clin Microbiol 1999, 37: 2543-2547.
Leparc-Goffart I, Baragatti M, Temmam S, Tuiskunen A, Moureau G, Charrel R, de Lamballerie X: Development and validation of real-time one-step reverse transcription-PCR for the detection and typing of dengue viruses. J Clin Virol 2009, 45: 61-66. 10.1016/j.jcv.2009.02.010
Lindegren G, Vene S, Lundkvist A, Falk KI: Optimized diagnosis of acute dengue fever in Swedish travelers by a combination of reverse transcription-PCR and immunoglobulin M detection. J Clin Microbiol 2005, 43: 2850-2855. 10.1128/JCM.43.6.2850-2855.2005
Paudel D, Jarman R, Limkittikul K, Klungthong C, Chamnanchanunt S, Nisalak A, Gibbons R, Chokejindachai W: Comparison of real-time SYBR green dengue assay with real-time taqman RT-PCR dengue assay and the conventional nested PCR for diagnosis of primary and secondary dengue infection. N Am J Med Sci 2011, 3: 478-485.
Sadon N, Delers A, Jarman RG, Klungthong C, Nisalak A, Gibbons RV, Vassilev V: A new quantitative RT-PCR method for sensitive detection of dengue virus in serum samples. J Virol Methods 2008, 153: 1-6. 10.1016/j.jviromet.2008.06.023
Dos Santos HWG, Poloni TRRS, Souza KP, Muller VDM, Tremeschin F, Nali LC, Fantinatti LR, Amarilla AA, Castro HLA, Nunes MR, Casseb SM, Vasconcelos PF, Badra SJ, Figueiredo LTM, Aquino VH: A simple one-step real-time RT-PCR for diagnosis of dengue virus infection. J Med Virol 2008, 80: 1426-1433. 10.1002/jmv.21203
Shu P-Y, Chang S-F, Kuo Y-C, Yueh Y-Y, Chien L-J, Sue C-L, Lin T-H, Huang J-H: Development of group- and serotype-specific one-step SYBR green I-based real-time reverse transcription-PCR assay for dengue virus. J Clin Microbiol 2003, 41: 2408-2416. 10.1128/JCM.41.6.2408-2416.2003
Waggoner JJ, Abeynayake J, Sahoo MK, Gresh L, Tellez Y, Gonzalez K, Ballesteros G, Pierro AM, Gaibani P, Guo FP, Sambri V, Balmaseda A, Karunaratne K, Harris E, Pinsky BA: Single-reaction, multiplex, real-time rt-PCR for the detection, quantitation, and serotyping of dengue viruses. PLoS Negl Trop Dis 2013, 7: e2116. 10.1371/journal.pntd.0002116
Waggoner JJ, Abeynayake J, Sahoo MK, Gresh L, Tellez Y, Gonzalez K, Ballesteros G, Balmaseda A, Karunaratne K, Harris E, Pinsky BA: Development of an internally controlled real-time reverse transcriptase PCR assay for pan-dengue virus detection and comparison of four molecular dengue virus detection assays. J Clin Microbiol 2013, 51: 2172-2181. 10.1128/JCM.00548-13
Warrilow D, Northill JA, Pyke A, Smith GA: Single rapid TaqMan fluorogenic probe based PCR assay that detects all four dengue serotypes. J Med Virol 2002, 66: 524-528. 10.1002/jmv.2176
Domingo C, Niedrig M, Teichmann A, Kaiser M, Rumer L, Jarman RG, Donoso-Mantke O: 2nd International external quality control assessment for the molecular diagnosis of dengue infections. PLoS Negl Trop Dis 2010.,4(10): doi:10.1371/journal.pntd.0000833
Levi JE, Tateno AF, Machado AF, Ramalho DC, de Souza VAUF, Guilarde AO, de Rezende Feres VC, Martelli CMT, Turchi MD, Siqueira JB Jr, Pannuti CS: Evaluation of a commercial real-time PCR kit for detection of dengue virus in samples collected during an outbreak in Goiania, Central Brazil, in 2005. J Clin Microbiol 2007, 45: 1893-1897. 10.1128/JCM.00065-07
Santiago GA, Vergne E, Quiles Y, Cosme J, Vazquez J, Medina JF, Medina F, Colón C, Margolis H, Muñoz-Jordán JL: Analytical and clinical performance of the CDC real time RT-PCR assay for detection and typing of dengue virus. PLoS Negl Trop Dis 2013, 7: e2311. 10.1371/journal.pntd.0002311
Waggoner JJ, Abeynayake J, Sahoo MK, Gresh L, Tellez Y, Gonzalez K, Ballesteros G, Guo FP, Balmaseda A, Karunaratne K, Harris E, Pinsky BA: Comparison of the FDA-approved CDC DENV-1-4 real-time reverse transcription-PCR with a laboratory-developed assay for dengue virus detection and serotyping. J Clin Microbiol 2013, 51: 3418-3420. 10.1128/JCM.01359-13
De Paula SO, de Melo LC, Torres MP, Pereira MRG, da Fonseca BA L: One-Step RT-PCR protocols improve the rate of dengue diagnosis compared to Two-Step RT-PCR approaches. J Clin Virol 2004, 30: 297-301. 10.1016/j.jcv.2003.11.004
Thai KTD, Phuong HL, Thanh Nga TT, Giao PT, Hung LQ, Van Nam N, Binh TQ, Simmons C, Farrar J, Hien TT, van Doorn HR, de Jong MD, de Vries PJ: Clinical, epidemiological and virological features of Dengue virus infections in Vietnamese patients presenting to primary care facilities with acute undifferentiated fever. J Infect 2010, 60: 229-237. 10.1016/j.jinf.2010.01.003
Lefrançois T, Blitvich BJ, Pradel J, Molia S, Vachiéry N, Martinez D: West Nile virus in Guadeloupe: introduction, spread, and decrease in circulation level: 2002–2005. Ann N Y Acad Sci 2006, 1081: 206-215. 10.1196/annals.1373.025
We are gratefull to Bénedicte Jos for excellent technical assistance.
The authors declare that they have no competing interests.
FN designed and conducted the data, FV performed the experiments and RC wrote the paper. All authors read and approved the final manuscript.